This Cerclage pessary is used to treat pregnant women with cervical incompetence in order to support the cervix and turn it towards the sacrum.
It may be indicated in pregnancies with a history of premature labour, multiple pregnancies or mothers who are exposed to physical strain (e.g. standing for a long time). It may also be indicated in pregnant women suffering from prolapse of the genital organs.
Using transvaginal sonography in recumbent or upright position we have established normal values for the cervical length and an opening of the cervical os (funnelling) for singleton and twin pregnancies. Low values of the cervical length and high values of the opening of the cervical os (Funneling) help identify patients who might benefit from pessary placement.
The pessary can easily be folded and inserted without pain. Before the application, it is recommended to exclude and treat if necessary cervical or vaginal infection. In general, cerclage pessaries should have a height of 17-25mm in singleton and a height of 25-30mm in twin or multiple pregnancies or pregnant patients with complaints of prolapse.
The width of the upper and lower diameter should be chosen depending on the individual constitution of the pregnant patient.
Perforated Cerclage pessaries are classified according to;
Application: The cerclage pessary is made of flexible comfortable, tissue tolerant silicone. Like all silicone pessaries it is elastic, and can thus be folded and inserted without pain. The cerclage pessary is designed for the treatment of pregnant women. It is to support the cervix especially by patients with additional complaints of prolapse (,,painful pressure downwards, mostly during standing and walking"), pregnant women who are exposed to physical strain (e.g. standing for a long time) or increased intrauterine pressure, or when an ultrasound shows signs of an incompetent cervix. Before insertion bacterial swabs should be performed to exclude bacterial colonization. The treatment should start early in pregnancy before the cervix might be effaced (at about 15-20 weeks).Transvaginal ultrasound at regular intervals can help to determine funneling of the internal os and detect the cervical length. The pessary is not meant to close the cervix but rather to support it and to shift the cervix towards the sacrum.
Sizes: Cerclage pessaries differ in their outer diameter (65mm to 70mm) as well as in the height of the curvature (each 17mm, 21mm, 25mm, 30mm). The inner diameter is either 32mm or 35mm. The higher models are preferred under more severe conditions.
Use: The cerclage pessary is inserted in the recumbent position by the obstetrician. It is vital that the curvature of the pessary faces upwards so that the larger diameter is supported by the pelvic floor. The pessary is folded and pushed into the vaginal fornix. While doing this, the cervix should be placed within the upper ring diameter. The obstetrician should
ontrol that the cervix is not surrounded too firmly by the upper ring of the pessary. After the insertion, the patient is asked to stand up and should hereby not feel the pessary when it is correctly placed. Possible prolapse complaints are directly redressed, which can motivate patient and team physician to continue with the treatment. The insertion of the pessary can be facilitated by applying a gliding compound, preferably an antibiotic cream preserving the natural flora. The pessary may stay until delivery, or at least until 37 weeks respectively. In cases of increased discharge it may be changed and cleaned in between, but this is not mandatory if manipulations are to be better avoided. Vaginal sonographic examinations can help identifying the cervical shape and length, the position of the pessary can easily be controlled manually during routine controls.
Side Effects/Complications: The right positioning of the pessary (with the curvature and the smaller diameter upwards) is the key to success of the treatment, otherwise the supporting effect is insufficient. There might be some increase of abacterial discharge, however, the spectrum of the vaginal flora will not be substantially altered.
Advice: The product should be used by one patient only. It can be stored by room temperature. The pessary can be cleaned under warm running water without using disinfectants.
Posted by pambam on 5th Apr 2011
My cerclage was extremely simple to use even with my bump :)
Posted by emmaonthehill on 31st Mar 2011
I accidentally ordered the wrong pessary for my Mum and MyPessary.com were quick to arrange a return and replacement order for which I am very grateful. Thank you and keep up the good work.